TENS & Muscle Stimulator MPO-8000 - Combo Professional Digital Unit
- Medical Products Online - Combo Professional Digital TENS & Muscle Stimulator System
- Lead Wires
- Four Self-Stick Hypoallergenic Disposable Electrodes
- 9V Battery
- Hard Carrying Case
- Safety Amplitude Cover
- Sturdy Metal Belt Clip
- Rubber side railings for maximum handling
- Instruction Manual
- 3 year warranty
DescriptionThe ease of use and versatility of the MPO 8000 Combo Professional Digital TENS and Muscle Stimulator System make this device professional in every way. Treat muscles and pain all with one elegant yet inexpensive unit. The reason many professionals rely on Combination units like the Medical Products Online 8000 is because of its reliability, versatility and more than ample output. This combination Tens Unit and Muscle Stimulator System provides professional treatment with portability and convenience all to help you feel better with each and every use.
Watch our VIDEO INSTRUCTION MANUAL
The MPO 8000 is a dual channel Digital TENS and Muscle Stimulator System with Five modes of TENS operation and Two Modes of Muscle Stimulator Operation. It is a battery operated Muscle Stimulator Unit that sends electrical impulses to the body to reach the nerves causing pain. The device is provided with two controllable output channels, each independent of each other. An electrode pair is connected to each output channel to transmit the electrical impulses to the area of pain.
The TENS operating modes include:
Strength Duration 1 (SD1)
Strength Duration 2 (SD2)
The Muscle Stimulator operating modes include:
The Medical Products Online 8000 includes a patient compliance meter to track treatment, and a battery indicator.
The Medical Products Online 8000 includes a timer so that you can easily set the treatment time. The electronics of the Medical Products Online 8000 create electrical impulses whose intensity, duration, number per second, modulation, pulse width, pulse rate, contraction, relaxation and ramp may be altered with easy to use digital controls. The cover prevents accidental changes in the settings.
You should discuss your treatment with your physician or therapist to achieve the greatest results.
Two Electric Muscle Stimulator Modes
Five Tens Unit Modes
Pulse Width: 30 300us
Pulse Rate: 1 - 150 Hz,
Forty-Five Adjustable Frequencies
On Ramp: Adjustable 1-6 seconds
Cycle ON time: Adjustable 1 - 30 seconds
Cycle OFF Time: Adjustable 1-30 seconds
Wave Form: Asymmetrical Bi-phasic square pulse
Timer: 1 - 60 minutes ( 1 min./step)
Patient Compliance Meter
Memory of the accumulated time from 0 to 999 minutes of executed applications
Technical DetailsTechnical Specifications:
Channels: Dual Channel, isolated between channels
Pulse Amplitude: Adjustable, 0-100 mA peak into 500 ohm load each channel.
Voltage: 0 - 50V (Load: 500 ohm)
Pulse Rate: Adjustable, from 2 to 150 Hz, 1 Hz/step
Pulse Width: Adjustable, from 50 to 300 microseconds, 10uS/step
Wave Form: Asymmetrical Bi-Phasic Square Pulse
On Time: Adjustable, 2~90 seconds , 1 Sec./ step
Off Time: Adjustable, 2~90 seconds , 1 Sec./ step
Ramp Time: Adjustable, 1~8 seconds, 1 Sec. / step, The "On" time will increase and decrease in the setting value.
3 EMS Modes: Constant , Synchronous, Alternate
Constant stimulation based on setting value. Only pulse width, pulse rate and timer are adjustable in this mode. "Constant" is equal to the "Normal" mode of a TENS unit.
Synchronous: Stimulation of both channels occurs synchronously. The "ON" time including "Ramp Up" and "Ramp Down" time. Therefore, the setting of ON Time should be no less than two times of the "Ramp" time in this mode. ON TIME � Ramp up + Ramp down
Alternate: The stimulation of the CH2 will occur after the 1st contraction of CH1 is completed. In this mode, the setting of ON Time should be no less than two times of the "Ramp" time. The OFF Time should be equal or more than the ON Time. ON TIME Ramp up + Ramp down OFF TIME ON.
5 Modes: B(Burst), N(Normal), M(Modulation), SD1(Strength Duration), SD2
Burst Mode: Burst rate: Adjustable, 0-5 - 5Hz; Pulse width adjustable, 50~300uS; Frequency fixed = 100 Hz
Normal Mode: The pulse rate and pulse width are adjustable. It generates continuous stimulation based on the setting value.
Modulation mode is a combination of pulse rate and pulse width modulation. The pulse rate and width are automatically varied in a cycle pattern. The pulse width is decreased by 50% from its original setting in 0.5 second, then the pulse rate is decreased by 50% from its original setting in 0.5 second. Total cycle time is 1 second. In this mode, pulse rate( 2-150Hz) and pulse width(50-300S) are fully adjustable.
The SD1 (Strength-Duration) mode consists of automatic modulation intensity and pulse width in 40% range. The intensity is always increasing while the pulse width is decreasing and vice-versa. The intensity is decreased by 40% while the pulse width is increased by 40% in 5 seconds. In the next 5 seconds, the intensity is increased by 40% while the pulse width is decreased by 40%. Total cycle time is 10 seconds. Pulse rate (from 2~150Hz) and pulse width (from 50~300S) are fully adjustable.
The SD2 (Strength-Duration) mode consists of automatic modulation intensity and pulse width in 70% range. The intensity is always increasing while the pulse width is decreasing and vice-versa. The intensity is decreased by 70% while the pulse width is increased by 70% in 5 seconds. In the next 5 seconds, the intensity is increased by 70% while the pulse width is decreased by 70%. Total cycle time is 10 seconds. Pulse rate (from 2~150Hz) and pulse width (from 50~300S) are fully adjustable.
Patient Compliance Meter: This unit can store 60 sets of operation records. Total recorded time is 999 hours.
Timer: Adjustable, from 5 to 60 minutes or Continuous. Adjustable in 5 minutes each step. Treatment time countdown automatically.