MPO-6300 Elite Digital TENS and EMS Unit
- Medical Products Online - Elite Digital TENS & EMS System
- AC Power Supply and 9-volt Battery
- Hard Plastic Case
- Two lead wires
- Four 2"x2" Square Cloth Premium Electrodes
- Metal Belt Clip
- Instruction Manual - Click here to download
- 2 year warranty
DescriptionThe MPO 6300 Elite Digital TENS and EMS Combination System is a fully digital combination TENS Unit and Muscle Stimulator System having has 17 modes available to the professional and the patient. Medical products online is excited to offer to patients and professional offices including orthopedic, chiropractic, physical therapy and pain management offices the most reliable high quality combination TENS and EMS Treatment System available anywhere. Combination units offer a wide variety of TENS and EMS Modes from the same unit!
Watch our VIDEO INSTRUCTION MANUAL
High power Digital TENS / EMS Combination Unit 100mA Strength
Digital controls allow 100 steps of function and intensity control.
Powered by either a 9V battery or the included AC adaptor
Power saving automatic turn-off feature if unit is not in use for 5 minutes.
The MPO 6300 Elite Digital TENS and EMS System provides five TENS modes, three EMS modes and nine preprogrammed treatment modes. This combination TENS Unit and EMS SYstem also features 100 mA output strength with 99 steps of intensity control, dual power from either a 9 volt battery or the included AC adapter. This TENS and EMS System also features a large LCD screen and user friendly buttons and a special pause function, which make repositioning electrodes so simple when treating multiple areas sequentially. There is also a power saving feature that automatically turns the TENS and EMS Unit off in five minutes when not in use. The fully digital format of this TENS device allows for reproducible and precise treatment formats for literally hundreds of painful musculoskeletal conditions.
The MPO 6300 provides the most powerful, effective and versatile portable Muscle Stimulator. Muscle Stimulation has long been one of the most difficult electrotherapy to provide until now. The stable Digital Programs offered by the MPO 6300 have been proven effective in the Real World of Pain Management and Muscle Rehabilitation. Whether the problem is Shoulder Rehabilitation, Hip Muscle Weakness, Post-Surgical Rehabilitation, Low Back Muscle Spasm, Whiplash Recovery, or other condition, the MPO 6300 will surpass any other unit in its class.
This TENS function of the MPO 6300 is primarily used for managing chronic pain. TENS devices work by sending dental pulsing electronic signals through the skin with electric connections that block pain signals coming from nearby nerves. In the TENS mode of operation, the MPO 6300 delivers mild electrical impulses to the skin to stimulate cutaneous (surface) and afferent nerves to help control pain. TENS does not have any known side effects unlike drugs or topical ointments or invasive pain management procedures.
Technical Details Channel: Dual, isolated between channels
Pulse Amplitude: Adjustable, 0-110 mA peak into 500 ohm load each channel.
Wave Form: Asymmetrical Bi-Phasic Square Pulse
Voltage: 0 to 55V (Load: 500 ohm)
Power source: One 9 Volt Battery and AC Power Supply.
Size: 11.8cm(L) x 6 cm(W) x 3.1cm(H)
Weight: 157 grams with battery.
Pulse Rate: Adjustable, from 2 to 150 Hz, 1 Hz/step
Pulse Width : Adjustable, from 50 to 300 microseconds, 10 μs/step
On Time: Adjustable, 2~90 seconds , 1 Sec./ step
Off Time: Adjustable, 2~90 seconds , 1 Sec./ step
Ramp Time: Adjustable, 1~8 seconds, 1 Sec./ step.
Mode: Six TENS Modes: B(Burst), N(Normal),M (Modulation Rate & Width),S1( Modulation Width), S2 (Modulation Width) and P Three EMS Modes:C(Constant), S (Synchronous), A(Alternate)
Burst Mode: Burst rate: Adjustable, 0.5 ~ 5Hz
Pulse width adjustable, 50~300 μs
Frequency fixed = 100 Hz
Normal Mode: The pulse rate and pulse width are adjustable. It generates continuous stimulation based on the setting value.
Modulation Mode: Modulation mode is a combination of pulse rate and pulse width modulation. The pulse rate and width are automatically varied in a cycle pattern. The pulse width is decreased by 50% from its original setting in 0.5 second, then the pulse rate is decreased by 50% from its original setting in 0.5 second. Total cycle time is 1 second. In this mode, pulse rate(2-150Hz) and pulse width(50-300μs) are fully adjustable.
S1 Mode: Pulse width is automatically varied in a cyclic pattern over a nominal 10 second period. Pulse width decreases over a period of 4 seconds from the initial setting to a value 40% less. The narrower pulse width continues for 1 second. It then increases over a period of 4 seconds to its initial setting. The cycle is then repeated. Pulse rate and pulse width are fully adjustable.
S2 Mode: Pulse width is automatically varied in a cyclic pattern over a nominal 10 second period. Pulse width decreases over a period of 4 seconds from the initial setting to a value 70% less. The narrower pulse width continues for 1 second. It then increases over a period of 4 seconds to its initial setting. The cycle is then repeated. Pulse rate and pulse width are fully adjustable.
Constant Mode(C): The pulse rate and pulse width are adjustable. It generates continuous stimulation is delivered .
Synchronous Mode(S): Output from both channels occurs synchronously. The "ON" time includes "Ramp Up" and "Ramp Down" time. Therefore, the setting of ON Time should be no less than two times of the "Ramp" time in this mode.
Alternate Mode(A): The stimulation of the CH2 will occur after Mode(A) the 1st contraction of CH1 is completed. In this mode, the setting of ON Time should be no less than two times of the "Ramp" time. The OFF Time should be equal to or greater than the ON Time.
Mode P: The pre-set parameters of the 9 programs are as given below:
Program Mode: 9 Preset Programs - Pulse Rate / Pulse Width / Timer
Timer: Adjustable, from 5 to 60 minutes minutes and continue(C), 5 minutes each step.
Patient Compliance Meter: This unit can store 60 sets of operation records. Total recorded time is 999 hours.
Low Battery Indicator: A low battery indicator will show up when the battery is low.
Operating Condition: Temperature:0℃~40℃
Relative Humidity: 30%~75%
Atmosphere Pressure : 700Hpa~1060Hpa